User name Eugene
Log entry time 11:50:34 on February10,2002
Entry number 82493
keyword=RCS HV wrong setting for Jan,9 - Jan,10
Today at 8:00 I discovered that a wrong HV settings have been used.
Searching the log files we found that a setting:
hv_setting_03 from Jan,25 was loaded on Feb,9 at 14:32, perhaps accidentally
These settings stayed till Feb,10 8:00
The newer (correct) settings have been loaded on Jan,10 at about 11:00.
The latest settings are stored in:
hv_setting_13 - text file (available with "File" command)
#30 - entry in the DataBase (available with "DataBase" command)
The HV software has been changed. It now gives more warnings
before actually changing the HV. Also, the hv_setting_ON/OFF
files are stored in a directory with only one (the latest) setting,
that it is more difficult to load an antique setting.