User name kuss
Log entry time 09:29:05 on March 7,2000
Entry number 35772
keyword=target density change
Lowered the target density from 155psi to 66.4psi. According to the diagram in the counting house this is a drop from 68mg/cm3 to 34mg/cm3, exactly 50% (this was pure luck).
Both fans are running at 25%, JT valve settings are 25% (JT1) and 30% (JT3). Both heaters together deposit 200W heat load, which is the equivalent of a 160µA beam current, at this density (1.2W/µA).
Target is ready.