User name sarty
Log entry time 05:56:39 on July 27, 2006
Entry number 175533
This entry is a followup to: 175513
keyword=Harp 6 stops working properly again...
Was going to try ensuring I had good cross-calibrations of (BPMs seen in
CODA using spot_L and spot_R) versus (harp scan), since I had just one
run earlier in the shift on spot_L. I figured I could do this with
raster on (checking the "extents" of the BPM monitoring) while Ron was
watching the heating curve on the 4% radiator with 4 uA. However, the
Harp 6 stopped working yet again. What happens is that it appears
"stuck" on the control screen (the little movement indicator stays still,
when it should be moving intiially, and then it starts moving but only
goes half-way). So, this data appears unusable.
A copy of this log entry has been emailed to: email@example.com