Make New Entry,
Make Followup Entry
User name R. Michaels
Log entry time 05:14:20 on March15,2009
Entry number 266226
This entry is a followup to: 266220
keyword=guess about DT bug
Per halog 266225, I think in run 2176 the deadtime is normal and
My guess about how the analyzer arrives at -22 % :
In that run, the L1A double-pulsing occured for ~40% of the data.
If the analysis only uses scalers for L1A, it would get an
avg DT = 0.40*(-70) + 0.60*15 = -19 %
Actually, the run 2176 should be checked by an expert.
The double-pulsing still should be fixed ...