Jefferson Lab Alignment Group

DATA TRANSMITTAL

TO: A. Saha, J. Gomez, J. LeRose DATE: 5 Dec, 2001 FROM: C.J. Curtis Checked: # : A720 DETAILS:

Below are the results from the Hadron spectrometer pointing carried out on November 27^{th}, 2001.

================ RESULTS =============== H112701A

The central ray of the spectrometer is at 28.289 degrees. It is missing the defined target center by 3.40 mm downstream, and -0.20 mm vertically (positive = up).

If the offset is corrected by secondary alignment, the spectrometer will be at 28.266 degrees.

*.9PR : 0.125 *.3DD : 1.18