Jefferson Lab Alignment Group

DATA TRANSMITTAL

TO: J. Arrington, J. Gomez, N. Liyanage, J. LeRose DATE: May 6, 2002 FROM: Chris Curtis Checked: # : A764 DETAILS:

Below are the results from the spectrometer pointing surveys carried out on the Electron (left) arm on April 30^{th}, 2002, and the Hadron (right) arm on May 1^{st}, 2002.

================ RESULTS =============== E043002A

The central ray of the spectrometer is at -22.155 degrees. It is missing the defined target center by 1.51 mm upstream, and 0.66 mm vertically (positive = up).

If the offset is corrected by secondary alignment, the spectrometer will be at -22.165 degrees.

*.9PR : 0.098 *.3DD : 1.12

================ RESULTS =============== H050102A

The central ray of the spectrometer is at 22.184 degrees. It is missing the defined target center by 3.20 mm downstream, and 0.15 mm vertically (positive = up).

If the offset is corrected by secondary alignment, the spectrometer will be at 22.162 degrees.

*.9PR : 0.137 *. 3DD : 1.67