Jefferson Lab Alignment Group

DATA TRANSMITTAL

TO: J. Gomez, J.P. Chen, N. Liyanage, J. LeRose DATE: 30 May 2001 FROM: C.J. Curtis Checked: # : A671 DETAILS:

Below are the results from the spectrometer pointing surveys carried out on the Hadron and Electron arms on May 24^{th} and 29^{th} respectively.

================ RESULTS =============== H052401A

The central ray of the spectrometer is at 19.998 degrees. It is missing the defined target center by 0.43 mm upstream, and - 0.10 mm vertically (positive = up).

If the offset is corrected by secondary alignment, the spectrometer will be at 20.001 degrees.

*.3dd = 1.94 *.9pr = 0.12

================ RESULTS =============== E052901A

The central ray of the spectrometer is at - 19.985 degrees. It is missing the defined target center by 2.29 mm upstream, and 0.47 mm vertically (positive = up).

If the offset is corrected by secondary alignment, the spectrometer will be at - 20.001 degrees.

*.3dd = 1.31 *.9pr = 0.11

Superharps DX DY locations in mm (+ is beam left)

SH1H01A - 0.04 0.18 SH1H01B 0.02 0.14 SH1H01C 0.02 0.14

SH1H02A - 0.11 0.15 SH1H02B - 0.05 0.18 SH1H02C - 0.03 0.22

2