User name slifer
Log entry time 20:55:33 on June12,2002
Entry number 85945
keyword=MCC update on Q3 problems
Richard from MCC calls to tell us that the recent right arm quad 3 failure
was caused by an inadvertant reboot of IOC HLA2. They were attempting
to reboot IOC HLC2 in hall C. The reboot apparently caught the JT valve
in closed state so the the magnet warmed up.