User name jpchen
Log entry time 19:32:19 on June21,2002
Entry number 86544
keyword=cryotarget cool-donw completed
Cryotarget cool-down completed at 19:30. Switched to cold-return around 18:30.
Loop1 is LD2 at 22K, ~27.0 psi. Since gas vented soem during warm up, we had to add some. The D2 gas bottle is empty now. Loop1 fan is set at 60 Hz.
Loop 2 is LH2 at 19K, ~25.8 psi. Fan is 24 Hz.
Vac gauge tripped off earlier, reset by Gary Dezern. It is reading ~5*10-7.