User name R. Michaels
Log entry time 01:48:50 on June22,2002
Entry number 86558
keyword=HARP problems and results
The previous shift noted that they could not analyze HARP scan data. We confirmed
that when one presses on the monticello button for "Hall A Scanners..." nothing
pops up. However, this GUI is working in MCC. So I went there and analyzed the
run 1798 which was saved by the last shift. Result was 59 x 83 microns (horizontal
by vertical at last BPM) which is a little on the small side but we think its ok.
Another problem is that we cannot find precise criteria for what is an acceptable
intrinsic spot size. The run plan says on page 9 that it should be "roughly 100
micron", so then what is the minimum and maximum allowed ? Say, not less
than ~60 and not more than ~150 ? This is what we assume.