User name Reitz
Log entry time 20:59:23 on October 4,2002
Entry number 87332
keyword=raster, bpm and espace
CODA runs 1131 and 1132 were taken in withroc14 configuration.
Raster was on. We've asked for 2*2mm and 1.5*1.5mm. spot_roc14
reconstructed 3*3mm and 2.2*2.2mm at the target position.
The otr target at 1h04a gave similar results.
The beamline ADCs in ROC 2 (upper crate, right HRS) do see
the bpm/raster information, the raw data looks fine (no saturation)
=> it is fine to run CODA in the coinc configuration.
However, at this time we have no absolute calibration for those ADCs,
since todays bulls eye scan failed - but one can do it later on.