User name paschke
Log entry time 16:05:43 on October25,2002
Entry number 89015
We were down all shift (except for first 13 minutes) due to G0 beam setup.
The new Q1 reset procedure has been tested. Q1 Left wouldn't come back up
after testing, so Scott and Mark worked on it all afternoon. I believe they replaced
the motherboard in the power supply, and it appears healthy now.
The cryotargets have been resting all day, with low coolant consumption. The G0 target was cooled down in the morning. In preparation for this, we set the JT valves for approx 200W HPH overhead on each loop, and which was plenty to ride out the slight coolant temperature burp.