User name Olivier
Log entry time 17:23:23 on July26,2003
Entry number 107101
keyword=No fast feedback
The FFB expert said there was a piece of hardware that was needed to fix the FFB, and they don't even have that piece. We basically have no other choice than running without it until monday.
So it is up to the operators, but even more to the shift workers, to carefully monitor the beam position, and to ask MCC to correct it as often as required. We want (0,0) with a tolerance of +-0.1 . There is a stripchart of the BPMs on adaqlr3 (the computer on the "angle" of the table in the counting house).