User name R. Michaels
Log entry time 13:32:01 on July29,2003
Entry number 107977
The lower CAMAC crate's power supply is broken and needs to be replaced.
It continuously blows fuse 16F (a 15 A, 250 V fuse), tried to replace it 3 times,
the supply makes an ominous noise and blows the fuse. Replacing with a spare
will take an hour. It was decided by RC to postpone this until tomorrow 8 AM.
Regarding the VDC HV, we reset the supply in the hut and (more importantly I
think) reset in the "Reset" GUI in the main EPICS Gui. The HV came on.
Instruction to shift: No point in running L-arm DAQ.