Make New Entry,
Make Followup Entry
User name V. Sulkosky
Log entry time 10:00:34 on February 04, 2009
Entry number 258904
keyword=Gas Cerenkov ADC #6 Status
I have looked at the data taken over night after I switched ADC delay
cables for ADC #6 for the left gas Cerenkov. The amplitude has been
stable overnight with no changes.
Figure 1 shows the pedestal subtracted amplitude for run 5697 (golden run)
versus one of the latest production runs, 5854. The single photoelectrons
peaks line up well and the main amplitude is also similar. All the
production runs from last night show the same behavior. I will keep on an
eye on it to make sure the problem is gone.
A copy of this log entry has been emailed to: bogdanw,jpchen,brads,jiang
ADC Number 6