A coincidence trigger is setup for the one-pass running. We had to get rid of about 60ns of hardware delay on T3(used by the CAMAC width/delay modules) at the BB weldment. That means we don't have remote adjustments on T3. And I effectively added 15ns to T1 trigger before going to coincidence. All this was done set the timing for the gates right, without adding extra delay on analogue signals. Re-timing on BB is still done with T1.
T1(EDTM pulse) comes 60ns after the coin. window(T3). T3 width is set to 140ns.
The edtm delay setting for this kinematics (proton ~870MeV) is ./edtmsetup 90 53
A copy of this log entry has been emailed to: email@example.com,firstname.lastname@example.org,email@example.com,firstname.lastname@example.org