Make New Entry,
Make Followup Entry
User name D. Parno
Log entry time 04:55:56 on February 14, 2009
Entry number 261084
This entry is a followup to: 261082
keyword=Compton vertical tune done
I've done a quick vertical scan and the new integrated field set point is
1116796.928 (corresponding to our IPM102B.YPOS = -0.93). This gave
good-looking rates at 10 uA during the scan; we'll have to see if they drift.
If the Compton HV (beamline card 11 slot 1) keeps tripping, you can turn
it down to -1750 V (and probably even -1700 V) if necessary. If the
Compton gamma rate disappears, you've gone too far.
A copy of this log entry has been emailed to: brads, camsonne