Make New Entry,
Make Followup Entry
User name brads
Log entry time 18:24:15 on February 26, 2009
Entry number 263538
This entry is a followup to: 263534
keyword=BB Cer PMT 18 TDC cable wasn't seated well -- fixed
BB Cer PMT 18 cable running to the TDC wasn't seated well at the 50 ohm
splitter -- fixed.
Signals on PMT 12 look good all the way up to the discriminator. There
is actually a little background that the y-axis auto-scaling is hiding on
the plot so there is actually some background present. We'll have to see
how it performs with beam.