Make New Entry,
Make Followup Entry
User name R. Michaels
Log entry time 03:44:29 on June 4,2010
Entry number 324401
keyword=Qsq runs, part 1
Runs 27721 and 6838 with VDCs on.
Pb/D tgt 3 for Qsq measurements.
Mistake: Quartz HV too low (set for integ. mode).
J. Wexler puts HV to 2000V.
Runs 6839 and 6840 are good.
Also runs 27722 and 27723.
On L-HRS I see the upper and lower quartz detector,
and on R-HRS I only see upper, yet it is not well
separated from pedestal. And no lower quartz.
Of course it could be due to software (?)