I did a quick checkout of the VDC for run 1095. Observations: - dead u1 wire group 23 - dead v2 wire group 1 - u1 wire group 22 seems reversed, rises instead of falls (see attached) - still one hot wire in u1 Also, I'm unclear as to the database detector map. Around lunch yesterday, Sergey went to great lengths to reverse the order of the wire groups (16 channels each) in the map, but around 15:30 yesterday, the map was switched back to the old one, which is used for replay now and gives very reasonable wire maps. So all that reversal business went out the window? Please log what's going on.