Make New Entry,
Make Followup Entry
User name pzhu
Log entry time 11:35:49 on March 13, 2012
Entry number 366212
keyword=changed attenuate for BPM and raster,connected helicity to ADC
Yesterday I found that BPM ADC will saturate in some case, I modified the
BPM attenuate . Also right arm slow raster is saturated above 1.5V, I
changed both arm's slow raster's attenuate(maximum accept 1.6V right now).
we need to do the harp scan again for this new configuration.
Also I connected helicity to ADC.
A copy of this log entry has been emailed to: firstname.lastname@example.org,email@example.com,firstname.lastname@example.org,email@example.com,firstname.lastname@example.org,email@example.com,firstname.lastname@example.org,email@example.com