User name P. Djawotho
Log entry time 21:43:59 on October 18,1998
Entry number 3151
The cell debris were cleaned up by RCG personnel and K. Slifer.
Q-curve of the pick-up coils yielded a peak value of 1/2 the old peak.
Still investigating the cause. Pick-up coils look ok: passed DMM continuity
test and registered 15.5 ohms on both coils.
Initiated a NMR water calibration.
New Helium 3 cell "Armageddon" will be installed upon satisfactory completion
of water calibration signal.