User name Templon
Log entry time 13:57:48 on June 2,2000
Entry number 42880
This entry is a followup to: 42828
keyword=vdc problem "solved"
Talked to Bogdan about VDC plots from last night. He indicated that
the problem has to do with a threshold too high on e-arm. When control
was lost, a 'reasonable value' was set locally before Hall was closed.
The value was reasonable but not optimal. Threshold should be lower.
Also explains resolution problem (but not why resolution is worse on V1
than on U1). Should call Bogdan to reduce e-arm threshold at next