User name R. Michaels
Log entry time 19:04:26 on October10,2000
Entry number 50034
Its a little hard to tell from home, but it looks
like after the cable fix the right PMTs on ADC
signals are all missing. Its important to run with
a relatively pure sample of T1 (right spectrometer trigger),
no T2's or other stuff, and look at ADCs. For example
from Sun or HP, I type
dataspy -R -s
and observe patterns of TDC, ADC where typically
the right PMT ADC is missing.
I guess someone has been flailing with the cables
and something is unplugged -- at the fastbus crate.
(I don't think I did it, but....)