User name R. Michaels
Log entry time 11:59:52 on November 1,2000
Entry number 51341
The scaler crate power supply problem was fixed.
So, there are now 10 scalers on the R-spectrometer.
There are 48 channels of ``normalization'' info, which
are split to three groups: 1) not gated by helicity, 2) gated
by plus helicity, 3) gated by negative helicity. To minimize
confusion, I tried to leave the old scaler arrangement alone
and only ADD scaler channels. The mapping is clear from
the scaler.config files in ~adaq/e99007/right/scaler and ~adaq/
e99007/left/scaler. Also see section on scalers in