User name Gomez
Log entry time 10:37:00 on November 8,2000
Entry number 51700
This entry is a followup to: 51673
keyword=Left HRS LVDT
The Left HRS Lvdt is not operational (reading ~1M, max range +/- 22500).
Internal settings of the Lvdt readout modules can be examined/changed
with the following MEDM screens:
./disp/HRSL/HacL_HPM_LVDT.adl (horiz - left hrs)
./disp/HRSL/HacL_VPM_LVDT.adl (vert - left hrs)
Same for right hrs except change "L" to "R"
non-sense values were found for the various fields of the
left hrs lvdt indicating memory corruption by radiation.
The fields were reset but the readings did not change.
>>>> Notify tech to cycle power of this unit in the next available