User name R. Michaels
Log entry time 12:01:18 on November28,2000
Entry number 52558
This entry is a followup to: 52527
keyword=trigger modifications (cont)
The trigger modifications on the Right Spectrometer are complete.
I have checked things out, looks ok. Some notes to add to halog 52527:
1. The "official" detmap in ~adaq/detmap/detmap.config (which is
inserted into the datastream and used by various online codes) was
updated to reflect the fact that the TDC signals for S2 plane on R-arm
2. New triggers on R-arm, more details
T1 = (S1-OR).AND.(S2-OR) ...... a simple overlap of S1,S2 planes
T2 = exclusive of T1, and require 2 out of 3 from S1,S2, Cerenkov
T7 = The old S-Ray trigger from MLU
I request to keep about 1 Hz of each of these for studies.
3. Related to 1 and 2, the xscaler's scaler.config file was updated.