User name K. McCormick
Log entry time 21:41:21 on January15,2002
Entry number 80347
The cryotarget sometimes stops half-way through a move (i.e. clicking to move on the "target lifting" GUI"). If this happens, the target will stop moving (i.e. the position will stop changing), the brake light will come on and the error light will come on. If this happens, click "Clear Select Record" (the second button on the lifter GUI), then re-choose the target you wanted and command the move again. This should make it complete the move. If not, reboot the IOC. This will reset the controls completely and you can complete the move.