User name Reitz
Log entry time 17:29:21 on January24,2002
Entry number 80585
Target vertical offsets were changed by 2 mm during the day.
When beam came back, beamspot at nominal beamposition (BPMB=-2.0/3.5)
was 2mm above the center, as it should.
New standard beamposition is:
**** -2.0 mm / +1.5mm **** (BPMBx / BPMBy)
at this position data was taken:
coda run 1602 / harp run 1482
results of harp scan: (using survey from Oct 2001, should be reanalysed with latest one)
1H03A x/y 1H03B x/y Target x/y
-531/504 1969/1015 3046/1235
asked for 4*4, resulted in 1.5mm*1.0mm on spot++ (coda run 1603)
asked for 5*5, resulted in 2.0mm*1.5mm (coda run 1604)