User name Rachek
Log entry time 19:11:28 on January31,2002
Entry number 81395
keyword=Broken filter-board replaced
Looking at gy:gx plot at calorimeter for high beam current runs
one could have seen "dark spots" - areas with substantiously
larger amount of events, although this is much less pronounced
if high cluster energy cut is applied.
Two most noticable regions are:
* block #137
* group of blocks which are summed in sum32 module #23
Today I have checked these channels.
* I found that there was a broken channel in filter board for
block #137 . That resulted in an incorrect shape of pulse
(it was abnormaly wide) at ADC for this channel and suppressed
pulse height which was overcome by increasing high voltage at
phototube. Another channel whose signal looks distorted is #135,
which is also in this filter board.
I have replaced the filter board by a spare one.
High voltages were reduced for channels #137 & #135
to get rouhgly the same laser pulse height as for other channels.
Calibration for lead glass blocks whose signals passed through
this filter-board should be checked/updated.
* Nothing special was seen in sum32->discriminator #23 line,
although a rate of double pulses at the output of this channel
looks larger than for other channels, if low threshold is set.
I changed discriminator channel (we have 8 spare channels in
one of the modules). It seems that this new channel has smaller rate of
double-pulses, although still a bit more than in other channels.