User name R. Michaels
Log entry time 12:02:27 on October 7,2002
Entry number 87561
This entry is a followup to: 87560
keyword=BPM pedestal run
For calibrating fastbus data I did a BPM pedestal run# 1204 in WITHROC14
config, so that as a sanity check I could verify the old ROC14 pedestals. They
looked fine, and so did the fastbus data. Calibration is in ~adaq/bpmcalib/calib.dat
The offsets and gains may still need small (~1-5 %) adjustments.
Run had IOCSE10 BPMs set to forced gain = 0. They are now restored to autogain.