User name paschke
Log entry time 10:17:39 on October25,2002
Entry number 89012
This entry is a followup to: 88940
keyword=Q1 trip: what to do
The Q1 reset procedure has been tested by Mark and Scott. Q1 Left is still not
entirely reset-able through software, but the good news is that the reset can now
be done without an access.
Q1 reset procedure:
1) on the Monticello Q1 power supply control screen, click the Safety/Reset
button. This should clear the interlock lights on the power supply front panel.
2) on this same interface, click the Power Off/Reset button. This should reset
the IOC. For the right arm, this should complete the reset, but step 3 will likely
be required for the left arm.
3) For the left arm, a hardware reset is probably still necessary. Press the
Left Q1 Emergency Stop button, and use the key (E-stop reset key, which is #4 in
the key box... and #1 in our hearts) to reset it.
4) Press the Power On button. It should now be possible to bring the magnet