User name J. Singh
Log entry time 06:58:30 on August19,2003
Entry number 111875
keyword=septum training time info
for future reference:
ramping the septum from 0A to 250A took 1 hr @ at ramp voltage of 115mV.
ramping the septum from 250A to 414A took 3 hrs @ a ramp voltage of 115mV.
ramping the septum from 414A to 427A took 20 min @ a ramp voltage of 120mV.
recovering (both coil temps 7K or less) from a septum training quench @ 427A took 7 hrs.
see fig 1. the black curve is septum read current