User name bogdanw
Log entry time 00:03:48 on December24,2003
Entry number 115129
the adjustment of HV was done by using the scripts
onlana.C, a1_calib.C, and a2_calib.C in the directory e94107/aero.
The updated db are in the same directory.
The single electron peak position adjusted to 100 (A1) and 50 (A2).
The sum of all PMTs(asum_c) has a peak at 600 for A1 and 1000 for A2.
These lead to 7.5 photo-electrons in A1 and ~ 20 in A2.