User name K. Wang
Log entry time 17:38:54 on January30,2005
Entry number 140368
keyword=Q trips and recover Left
Q1, Q3 of HRSL are tripped off
stop run 2199
Bogdan is doing diagnostics
Level of LN2 an He dropped 1 minute ago
16:30 Controll access, Bogdan needs to work in the Hall
16:45 Try Q3 power back on when 'Quench OK', but the power
can not be on. Robert goes inside the Hall.
16:58 By changing p0 setting, Q3 current is going up.
17:00 Q1 is powering back.
17:10 Q3 is back to setting
17:20 Q1 is back normal, then dipola is back
17:25 ask for beam permit
17:35 MCC notice beam being ready
17:37 start run 2200