User name Higinbotham
Log entry time 12:42:49 on March10,2005
Entry number 141088
keyword=Comment on Gain Settings
This morning the gains look nicely match, though due to an interesting bug in the calibration code the dE is at a much high gain than the E.
This was due to the fact there are two different dE energies possible
for a single E cut. (see figure)
dE | *
After looking at the data we decided to just increase the threshold
of the dE as the gains were indeed well matched. We did lower the
HV of all of the dE's by 50 V.