User name R. Michaels
Log entry time 09:37:34 on July27,2005
Entry number 149647
This entry is a followup to: 149427
keyword=cavities saturating, Musson adjusts gain
The (new) XYQ cavities were saturating in their position
measurements for large raster size. This explains the bedposts
in halog 149427. Musson sees this on his own DAQ (the PC104).
So, this morning at 09:30, Musson adjusts the gain to get rid of
the bedposts in PC104 DAQ. We still need to execute the
commissioning run plan but this was clearly a necessary adjustment.