User name jpchen
Log entry time 12:35:04 on July21,2006
Entry number 174722
keyword=Fan speed at 48 Hz
Fans in both loops put large amount of heat into the loop. I had to
increase coolant flow significnatly to be able to get to 60 Hz.
I decreased both loop to 48 Hz (20%), the heat load reduction is
more than 100 W. So for now leave both at 48 Hz. Reduced coolant flow
to keep HP heater around 50 W. ESR flow is ~7 g/s (reading 10 g/s - offset of ~3g/s).