User name R. Michaels
Log entry time 10:44:01 on April25,2007
Entry number 199991
keyword=swapped CAMAC delay to fix missing TDC
I swapped out the CAMAC delay on L-arm CAMAC crate#1, slot 8.
This morning I observed that there was no output from this
module for S1-R-5th. The module which feeds this is the
ECL Progr. Width Module in slot 5, and it had all channels
functioning. One can conclude that the problem was either with
the cable or the module in slot 8 or its connector. Rob F tried
swapping the cable last night but the signal went away after
he left. The cable that's there now looks fine to me.
If it goes bad again then I think the only possibility is
the output connector on the ECL Progr. WM in slot 5. Doubt it.
One other thing: the trigger was NOT affected, only the TDC info.