User name sarty
Log entry time 16:31:45 on June13,2007
Entry number 205368
keyword=HV for HRS-L Pion Rejector 2: turned off
LeCroy HV channels tripped off, and would not turn back on (for some reason). The channels are all indicated to be connected to the HRS-L "Pion Rejector 2 (PRL2)", which we are not using for this experiment.
Therefore, the affected LeCroy HV channels were simply turned off. The attached screen shots show the channels turned off (they are in red) - the nominal set-values for these channels can also be seen from the screen shots (in case somebody wants to turn them back on later).