Make New Entry,
Make Followup Entry
User name Chiranjib
Log entry time 07:52:36 on November 1,2008
Entry number 245867
keyword=Couple of issues
1.During our NMR measurments, the wavetek 80 that provides the trigger to the lockin tripped a lot ( in the NMR field sweep VI ). We rebooted each time it happened.
3. J P mentioned before he left that if we are getting good increasing polarization , we should just let it pump and polarize till we get to the point of "polarized he-3 longitudinal" data taking
2.Just to keep a record, the output of the photodiode during all the EPR measurements is around 150 mV.