Make New Entry,
Make Followup Entry
User name D. Parno
Log entry time 12:19:42 on November 15, 2008
Entry number 248993
keyword=Scaler problem fixed (new Compton photon DAQ)
We have fixed a problem with the Caen V560 scaler in the VME crate with
the new Compton photon DAQ. The scaler had not been clearing properly
(instead of going to zero, each channel went to 25890 when reset).
We identified the problem as involving fixed bits. We moved the scaler to
a different slot in the VME crate, making sure that it was firmly in
place. The scaler channels now correctly reset to zero.
A copy of this log entry has been emailed to: firstname.lastname@example.org, email@example.com