Make New Entry,
Make Followup Entry
User name Moffit
Log entry time 12:01:59 on November 18, 2008
Entry number 249540
keyword=HAPPEX DAQ - Moved x10 BCM to adc15_1
x10 BCM (bcm4) signal was saturating adc22_3. So I moved it to adc15_1.
Signal at 12uA sits around 30k ADC channels. I suspect they will run up
to 15uA... so we should be good from here on out.
This occurred during run 30419. The database was also changed but will
not take into effect until the next run (30420) is started.
After the pedestal is calibrated, we'll use this bcm for charge asymmetry
A copy of this log entry has been emailed to: rom,mcnulty,mercado