Make New Entry,
Make Followup Entry
User name R. Michaels
Log entry time 04:39:54 on August25,2009
Entry number 285796
keyword=Notes on access
I'm assuming John Wexler is making an halog about the
PREX detector work. Here are some additional notes:
We temporarily used the PREX detector for trigger T3.
Using that we could see a good gate timing at the ADC.
Using the normal gate timing or the S0 trigger we can
see hardly any correlation. I think the gate timing is ok.
There is ~1 kHz of noise coming from this detector with
threshold like ~20 mV. Maybe it's ok.
Asked for beam permit at about 4:10.