Make New Entry,
Make Followup Entry
User name mercado
Log entry time 10:56:19 on August28,2009
Entry number 286551
keyword=Calibrating BCMs for 100uA
In preparation for 100uA beam, I moved around some of the BCMs in order to get
healthy signals from our 16bit ADCs in the Counting House.
BCMs 2 and 5 were moved:
BCM2 was in 22_1. Moved to 15_0.
BCM5 was in 2_1. Moved to 22_1.
All boards should be run at LO gain.
Board 22 MUST be on HI gain (~40000 signal for 100uA, as opposed to ~10000 in