Shift start with control access. LHRS is ready. Vince and Ramesh are working in RHRS. RHRS diple is ramping up. RHRS momentum setting is 2.36GeV. VDC off on both arm. Empty target.
16:22 Reset RHRS momentum to 2.63GeV/c; Cycling Q2/3
16:50 MCC called that they are ready to tune beam through Hall A. Asking them to tune the beam to nominal beam position on white board: BPMA = (-2.0, 1.25)mm BPMB = (-3.0, 1.5)mm
17:30 MCC can not get beam to requested position due to compton is now in beam. What we get is BPMA = (0.3,1.3) BPMB = (1.4,1.5). Continue to Carbon with hole target.
17:33 Start detector check out run: 25269/4814
17:40 Carbon hole data tell us we are far off on beam X position as we expected. Detector check out on both arm is on going.
17:48 Continue to moller measurement
22:04 moller measurement done: Halog 297889
22:14 BPM -> BPMA = (-2.0, 1.25)mm BPMB = (-3.0, 1.5)mm after MCC applied some magic .....
22:17 Target -> Carbon hole to confirm beam position. Run: 25271/4816 ........
22:41 Established our golden beam position BPMA = BPMB = (-3.0, 2.0)mm, which perfectly center carbon hole and without beam angles.
22:42 Harp scan again. There are Problems (Halog 297918). But we decide to move on to Q^2 measurements
23:00 Q^2 Measurement : LD2 Target, Taking run at both L/R HRS. VDC on. >200k Event each
23:45 Try to deal with a BPM and high dead time problem on LHRS. See Halog 297942, 297946, 297948. Bob called in to help solve fix the DAQ.