Make New Entry,
Make Followup Entry
User name R. Michaels
Log entry time 10:46:05 on November 07, 2009
Entry number 298074
This entry is a followup to: 298038
keyword=L-HRS RT module swapped out. put in spare
To effect a more permanent fix to the problem of flakey connector
on the RT module, I swapped it out on the L-HRS. I will get the
old one fixed by the electronics group.
I checked the delay overhead -- it's 120 nsec. It's normal/good.
This means if there is no RT relative to L1A the timing jumps
by 120 nsec; you would see this in VDCs as data to the right of
the normal TDC spectrum if there was an associated VDC hit.