Make New Entry,
Make Followup Entry
User name LeRose
Log entry time 18:30:23 on July 1,2010
Entry number 329315
keyword=Right Arm Set
The right HRS is now set for 1.131 GeV/c
There were problems getting the NMR to lock for the right dipole. The solution was to, after typing in the desired P0, change the NMR settings on the NMR control screen as follows.
Select probe A
set the compensating coil voltage to 4.0 Volts (the original setting is too low).
See the pictures attached