Make New Entry,
Make Followup Entry
User name Bob / Bogdan
Log entry time 23:25:33 on July 1,2010
Entry number 329357
keyword=R-HRS gate / timing adjustment
Bogdan adjusted the timing of the ADC gates on R-HRS.
Front panel gates were enabled on all 5 ADCs in the system, but
there was a glitch: ROC1 slot 24 does not see it's front panel
gate. I verified by swapping cables around that the gate is
definitely there, but the module does not see it. However it
does see the back panel gate and that's how I'll leave it overnight
so that the rest of the crate is happy. I'll swap out this ADC
Except for that glitch, I think the trigger and DAQ are finished.