The left q1 is up and the current is stable at 1.131 Gev. To take cabling issues out of the equation the control card extension cabling has been removed. However, this did not alleviate the oscillations. We replaced the DAC with the spare and the magnet miss-match issue went away. Additionally, the output set point and the read back were better aligned. The current fluctuations are now within acceptable tolerances. Will continue to monitor.